Название файла: DVP-28SV_instruction_en.pdf
Размер файла: 393 Килобайт
Количество загрузок: 1065
Pilot tests were also performed in different days so as to adjust the process. This becomes a powerful tool towards patient safety. When reviewing each CBC, it was observed that the reason for retention was not clinically relevant and it was decided, therefore, to re-evaluate the criteria for inclusion in the group of autoverified results. Before starting the autoverification process of 16,884 automated CBC per month, the NCP percentage was almost 0.065%, and manual typing of the differential leukocyte count considered the main reason for errors. Tool continuous improvement will allow the performance continues to rise. Testes piloto foram realizados em dias diferentes para ajustar o processo.